Love the nano data logger, just trying it for the first time -- noticed the first elevation readi...
Public Lab is an open community which collaboratively develops accessible, open source, Do-It-Yourself technologies for investigating local environmental health and justice issues.
This is a testing site only. See the live Public Lab site here »
All topics »
If you cannot use the ReCaptcha to verify you are not a bot, use this alternative verification.
As an open source community, we believe in open licensing of content so that other members of the community can leverage your work legally -- with attribution, of course. By joining the Public Lab site, you agree to release the content you post here under a Creative Commons Attribution Sharealike license, and the hardware designs you post under the CERN Open Hardware License 1.1 (full text). This has the added benefit that others must share their improvements in turn with you.
sign up to join the Public Lab community
Forgot your password? Reset it here
by liz |
June 24, 2017 19:34 |
Love the nano data logger, just trying it for the first time -- noticed the first elevation reading keeps showing up 3675.56. Any reason why?
Way to go getting the Nano Logger logging! The first elevation data point is always wacky because the first barometric pressure data point is always wacky. Elevation is computed by the sketch directly from pressure (hPa). Note that the first temperature data point is also erroneous and always the same.
I don't know whether these first data points are default values inserted by the library when no real data are available or whether they are more random. I do know that they should be ignored.
Don't forget to edit the sketch so that the current sea level barometric pressure is entered before you use the logger to save real elevation data. You can do the conversion later from the hPa data if you know the formula, but you have to know what the barometric pressure was when the data were collected. Get the sketch here.
Reply to this comment...
Log in to comment
Here is a better answer. The first time the sketch tries to read the BMP280 sensor is immediately after the sensor has been initialized. So the sensor is not ready and erroneous data is read.
So I put a 0.5 second delay after the initialization of the sensor and that seemed to fix the issue.
The updated sketch is available at the Guides page and here.
This is part of: