Dear public lab contributors, fascinated by a simple cardboard-spectrometer (based on a CD as di...
Public Lab is an open community which collaboratively develops accessible, open source, Do-It-Yourself technologies for investigating local environmental health and justice issues.
By signing up, you agree to the Code of Conduct, which applies to all online and in-person spaces managed by the Public Lab community and non-profit.
As an open source community, we believe in open licensing of content so that other members of the community can leverage your work legally -- with attribution, of course. By joining the Public Lab site, you agree to release the content you post here under a Creative Commons Attribution Sharealike license, and the hardware designs you post under the CERN Open Hardware License 1.1 (full text). This has the added benefit that others must share their improvements in turn with you.
Dear public lab contributors, fascinated by a simple cardboard-spectrometer (based on a CD as diffraction-grating) I wonder if it's possible (and affordable) to construct a DIY-spectrometer which can be used for measuring chemical concentrations. In particular it's about analyzing soil for ordinary garden plants with regard to nutritient content.
I would be glad about any clues /hints about what kind of diffraction-grating, what type camera etc. is needed/sufficient for that.
@stoft and/or @gretchengehrke, i'm curious about your input on this -- does the beer lambert law hold for opacity of a reagent? I guess it depends on if the reagent color change is linearly related to the concentration of the target compound (nitrate).
If it does, this is another reason to push our #cuvette-frame design forward, as we'd likely be able to do a similar photometric analysis of the reagent color change using either a fixed-exposure camera or two or more standard control samples at known nitrate concentrations.
It is my understanding that Beer's law is not the relationship between reagent's effect on color, but is the property of light passing through a uniform substance and the attenuation of that light with distance -- which is linear assuming the substance is uniform. So, if you had some chemical reaction, which was not linear with the concentration of reagent, that non-linearity would appear in a plot of the measured attenuation (from the spectral data) as the concentration changed. The Beer's Law linearity you are relying on is that given that the sample containers are all "identical" then the path-length for light is "identical" and therefore (with all else held constant - like the light source, camera exposure, etc) then the measurements of relative intensity will be directly related (linear or non-linear) with the concentration. [Note: Realize that there are many possible components to the issue of measurement stability and that the default PLab stuff is only a first step. You might find some of my PLab notes on spectrometer stability, noise, time-averaging, calibration and uncertainty of interest.] It's all in the numbers to estimate measurement limits vs your requirements.
Reply to this comment...
I think that makes sense. I didn't mean Beer Lambert would be related to the reagent's affect on the color, but merely that if the reagent reacts linearly with the substance, then Beer Lambert would ensure (all else being equal, as you point out) linearity of transmission of light through the reagent-reacted substance. But controlling all the other variables is not simple!
Thanks a lot for your advice, sorry for my late reply.
Since I regard myself as spectrometry-newbie reading the comments felt a bit like “biting off more than one can chew”, however the plan is complete / ready now: At first I'll learn about that colorimetric reagents (which I hadn't in mind before, thinking measuring mere reflection-effects would work as well). Secondly I'll probably learn about Beer's law or similar concentration measurement principles. (The nutritients I'm interested in measuring at the moment are in particular: Nitrogen, potassium, phosphorus but in the long run I'm eager to see which other nutritients can be spectrometrically measured).
Of course I'll also have a detailed look at the elaborate ready to use devices that were mentioned above for they seem to be a good alternative.
Thanks again for sharing your knowledge and thus jump-starting my hobby-project.