I finally opened the Spectronic 20 underside and here's what it looks like! Great to see they jus...
Public Lab is an open community which collaboratively develops accessible, open source, Do-It-Yourself technologies for investigating local environmental health and justice issues.
This is a testing site only. See the live Public Lab site here »
If you cannot use the ReCaptcha to verify you are not a bot, use this alternative verification.
By signing up, you agree to the Code of Conduct, which applies to all online and in-person spaces managed by the Public Lab community and non-profit.
As an open source community, we believe in open licensing of content so that other members of the community can leverage your work legally -- with attribution, of course. By joining the Public Lab site, you agree to release the content you post here under a Creative Commons Attribution Sharealike license, and the hardware designs you post under the CERN Open Hardware License 1.1 (full text). This has the added benefit that others must share their improvements in turn with you.
sign up to join the Public Lab community
Forgot your password? Reset it here
I finally opened the Spectronic 20 underside and here's what it looks like! Great to see they just use an incandescent bulb.
Can we connect it to an Arduino?
and how much does that thing cost Jeff ?
Is this a question? Click here to post it to the Questions page.
Reply to this comment...
Log in to comment
I see similar models for <$200 on ebay:
but of course that's because it's >30 years old and you have to read out the wavelengths one at a time by turning a dial :-)
Classy! On Ebay, you can also see the brand name change as the company was presumably acquired by different parent companies. I believe it eventually became part of ThermoFisher Scientific.
The optical diagram for the Spectronic 20 from here. It has two lenses.
Below, the optical diagram for the Spectronic 20+ from the official manual. It has three lenses.
It finally dawned on me that this machine is a monochronomater, attenuater, and photometer. To use it that way all you need to do is to bring out a portion of the beam with a 90/10 mirror and let the 10% hit the photometer. To do this just modfy a cuvette by adding a 90/10 mirror at a 45 degree angle. If you can find some multimode fiberoptic cable and mount it above the mirror and you are in business. Otherwise you can mount the public lab spectrometer above the modified cuvette and "go to town" with the setup.
Wait, you think that it hits the sample with a monochromatic light source? I guess we could easily test that with a mirror, as you describe. That'd be really neat. I just assumed it was diffracting after light passed through the sample, but with their setup, there's no reason they couldn't do it before.
Dan's right that it's a monochromator in that it directs light of one color (a narrow band of wavelengths) through the sample and to the detector. You then write down the intensity of light that hits the detector, and turn the dial to a different color. The dial mechanically turns a cam that rotates the grating so a different color is aimed at the exit slit. The instructions say to use 25 nm increments, which suggests that the resolution is not much better than that.
So I can't spell and the spell check did not help much.
Another thing to do is just appropriately scratch a small area on a fully reflecting mirror so some of the light gets to the meter in order to get a relative intensity value.
I also have an idea on how to effectively couple the light into the fiber if you need to go that route.
Login to comment.
This is part of: